The family of a man who bled to death during kidney dialysis treatment at Royal Shrewsbury Hospital have said they believe lessons have been learned.
Mohammed Ismael Zaman, known as Bolly, died after hospital staff failed to check the connection on his dialysis machine, despite it sounding an alarm after the catheter had become disconnected.
During Mr Zaman’s treatment at the Royal Shrewsbury Hospital on October 18, 2019, his dialysis machine set off a venous pressure alarm.
An unidentified member of staff reset the alarm without checking that the connection was still secure. As a result of the reset, Mr Zaman bled out for seven minutes losing 49% of his blood circulating volume.
He was found unconscious in a pool of blood and despite resuscitation attempts, died two hours later.
The coroner, Mr John Ellery concluded that the death was due to systems failure and individual neglect on the part of the unidentified staff member.
Read full story
Source: Shropshire Star, 16 January 2021